The device is designed for self-measurement of blood glucose. The Accu-Chek Performa glucometer is characterized by ease of use, speed of measurement and precision of the result. The device has a memory of 500 measurements.
The Accu-Chek Performa Glucometer Included:
Glucometer, case, lancing device and one barrel with 6 Accu-Chek FastClix lancets, Accu-Chek Performa test strips - 10pcs, battery, user manual, quick guide.
How the Accu-Chek Performa Glucometer works:
Testing blood glucose with the Accu-Chek Performa Glucometer is very simple, making it suitable for self-monitoring of blood glucose. The device meets strict standards and ensures measurement accuracy. The convenience is a readable screen, memory function and calculation of average measurements as well as test strips that quickly draw a drop of blood.
Accu-Chek Performa glucometer recommended for use in order to independently perform blood glucose measurements.
Store the product out of reach of children at room temperature. It is recommended to protect against the effects of moisture and light.
Display: large LCD screen. Power supply: 1 battery, type CR 2032. Battery consumption: approx. 1000 measurements. Blood drop size: 0.6uL, measurement range: 10mg / dL-600mg / dL. Blood sample type: capillary, venous, arterial and neonatal. Ability to measure from alternative sites (AST): YES. Hematocrit Range: 10 to 65%. Measurement time: about 5 seconds. Measurement temperature: air temperature: 8 ° C-44 ° C. storage temperature: air temperature: -25 ° C-70 ° C. Low blood drop warning: YES. Memory capacity: 500 results with date and time and 20 control test results. Dimensions: 94mm x 52mm x 21mm. Weight: 59g (with battery). Data transfer to computer: via infrared port and Accu-Chek Smart Pix device (available separately). Measurement principle: electrochemical measurement. Calculation of average results: from 7,14, 30 and 90 days.
How to use the Accu-Chek Performa glucometer:
Use according to the manufacturer's instructions. Read the instruction manual before use. Before the examination, wash and dry your hands. Insert the test strip into the device. Place your pricked finger with the blood sample on the side with the yellow window. The result will appear on the screen after 5 seconds. After removing the test strip and after 5 seconds, the device will turn itself off.